Checklist for schedule-based observations with SARDARA
Notice that actions take place in three different “locations”:
(1) = action to be performed in a terminal on the observing machine (the one with the DISCOS system interface)
(2) = action to be performed in a terminal on the project/schedule/data machine
(op) = command to be given in the operatorInput panel of DISCOS
Login on both (1) and (2) Connect via VNC to (1) as “observer”, from there use ssh to access (2) with your project-specific account.
Launch the monitors, if necessary (1):
Set the project name (op):
Initial setup (op):
> setupKKC (or other receiver code: CCC, XXP)
Tune the local oscillator, if any (op):
> setLO=[freq] —> e.g. setLO=22000 - start frequency of the observed band will depend on the backend
Point the antenna to a reference position (op):
> goTo=[Az]d,[El]d —> e.g. goTo=180d,45d
Always explicitly choose the Total Power backend (XARCOS might have been left active) (op):
Measure the cold sky signal level (op):
> getTpi > setAttenuation=[sect],[att] (values range from 0 dB to 15 dB, 1-dB step) —> iteratively adjust attenuations until the level is about 700-900 counts
Get a Tsys (op):
Pointing check (op):
> track=name (choose a proper calibrator from source catalogue) > chooseRecorder=MANAGEMENT/CalibrationTool —> the following command on (1): $ calibrationtoolclient (to display the plots) > crossScan=HOR,0.2d,00:00:15 (set proper parameters according to your beamsize) > azelOffsets=0d,0d (only if wanting to reject the measured offsets!)
System offsets, such as the ones measured with a Point acquisition, sum up to the ones indicated inside schedules ONLY if they are expressed in the same coordinate frame.
This means that, if you perform observations using EQ offsets, also the fine-pointing cross-scans must be carried out in the EQ frame. The same holds for HOR scans. If there is a frame mismatch, the system offsets are automatically rejected (bug under fixing).
Choose and set the spectrometer (op):
> chooseBackend=Sardara > initialize=[code] (e.g. SCC00, SCC00S, SK00, SK00S)
If non-default parameters are needed, configure the sections (op):
Remember: for full-Stokes setups, only section 0 is to be configured. For spectral-only setups, both sections 0 and 1 must be configured.
If a non-default value is needed, set the integration (op):
> integration=[n] (expressed in milliseconds)
Measure RMS in order to verify the signal level (op):
Values should be included in the 20-22 range. If they are not, adjust the attenuation levels as follows and repeat the RMS measurement.
If needed, set the attenuators (op):
Each attenuator can be set from 0 dB to 15 dB, with a 1-dB step.
If needed, create a schedule (2):
Use schedulecreator (see its own guide): $ basie –c [configfile] [out_directory]
Parse the schedule (2):
$ scheduleChecker [schedulename].scd —> Move the schedule files to the observing machine
Launch the schedule (op):
The project ID written inside the .scd schedule file MUST coincide with the one invoked during the observations (and thus with the project-specific path it is stored into).
Use the online quick-look provided on the second desktop of the remote machine.
Weather conditions and webcam (in a web browser)
Weather: www.med.ira.inaf.it/escs/meteo Webcam: www.med.ira.inaf.it/webcam.html
Stop the schedule (op):
- Copy the data (2)
—> Get the latest subfolders written in the main data folder
Stow the antenna and restore default user (op):
> antennaPark > project=staff